- Biotech degrees / certificates
- Event Materials
- Photo Gallery
Microtitre Plate Reader
SOP for DuPont Microtitre Plate Reader:
The microtitre plate reader is used to measure absorbency by measuring the amount of color against light. The microtitre plate reader is used when determining results from the TintElize immunoassay.
- Turn the machine on using the On/Off switch on the right hand side of the machine.
- The machine should be set at 492 nm.
- Place 96 well plate on machine, with one well under the nozzle.
- Press "measure".